Medical Equipments Mobile Energy Recovery ICU Anesthesia Ventilator
How does our ICU ventilator work?
The MCB-PA-700B ventilator is gas driven by medical compressed oxygen and compressed air. In the inspiratory phase, two streams of compressed gas (compressed oxygen and compressed air) flow into a high performance air-oxygen mixer to form a mix of oxygen and air with a certain pressure.
What is the detail of our mobie ventilator?
Display: 10.4''LCD Display Following Parameters & Real time waveform display
1.Operating mode and respiratory rate;
2.Ventilator control frequency set by medical staff;
3.Tidal volume of each breath in the patient;
4.Occurrence of spontaneous breath in the patient and spontaneous breathing frequency;
5.Expiratory phase and inspiratory phase and actual respiratory rate;
7.Setting and adjustment of inspiration triggering pressure, PEEP, and highest airway pressure;
8.Real-time variation in internal airway pressure.
1.Environmental Protection Material construction, hardly to be broken.
2.Trolly: Fitted with 4 nos anti-static rubber castors; two of which are lockable for braking and easy maneuverability with foot operated brake provisions.
1.Power: AC220V + 22 v, 50 Hz + 1 Hz and 300 va has a good reserve of single-phase ac power supply
one deep breath per 80 controlled respirations, the inspiration time is 1.5 times of the setting point
Pressure limit range
Airway pressure alarm: Audible and visual and with yellow and red color indicating
Lower: 0.1 kPa ~ 6.0 kPa; Upper: 0 ~ 5.0 kPa
Ventilation Volume alarm:
≥ 18L/ MIN
Power Supply Alarm
Ac/dc power supply are after failing to send out alarm immediately Alarm Time: keep >120s
The airway pressure continues to be higher than 15 hPa ±1 hPa for 15s±1s, then the machine will raise an audible alarm, the pressure will be displayed in red and the continuous high pressure red alarm wording is displayed on the screen of the anesthetic respirator.
10 ~ 40oC
no higher than 80 %
860 hPa ~ 1060 hPa
100-120 Vac, 50/60 Hz;
Attention: the AC power supply used for the anesthesia machine must be well grounded.
Attention: the anesthesia machine used must be equipped with a carbon dioxide monitor complying with ISO 9918:1993, an oxygen monitor complying with ISO 7767:1997 and an expiratory gas volume monitor complying with 18.104.22.168 of Medical Electrical Equipment Part II: Special Requirements for Safety and Basic Performance of Anesthesia System.
-15oC ~ +50oC
no higher than 95 %
86 kPa ~ 106 kPa.
It should be stored in a room without corrosive gas and well ventilated
comply with the requirement of GB/T 15464
Between the packaging box and product, soft material with appropriate thickness provided to prevent loosening and mutual friction during transportation
Moisture protection and rain protection to ensure that product is protected from natural damage.